Quick Links. Table of Contents. Page 2: Table Of Contents Duct adapter for remote sensor Explosion hazard! Not suitable for use in oxygen-enriched atmospheres, i. Page 6 — Special calibration mode blocking of alarm triggering, display of calibration mode, one-man calibration.
|Published (Last):||10 February 2019|
|PDF File Size:||6.3 Mb|
|ePub File Size:||4.58 Mb|
|Price:||Free* [*Free Regsitration Required]|
Docking Station Part No. Keep it in a safe place and refit it when installation. Cut excess wires short. Ex-Schutz Kennzeichnung vom Transmitter entfernen. Cut excess wires short or. If installing the transmitter in Zone 2 explosion-hazard areas or in areas not exposed to explosion hazard:.
Bei dem Einbau des Pumpenmoduls muss das Polytron von der Spannungsversorgung. Then insert. Clean if necessary. The opening for the eccentric catches must face upwards, engaged position.
Attention: use only an Allen key 5 mm without golfball. The resistance increases slightly approx. Note: If the connector fails to engage correctly, check that the terminals in the docking station are correctly aligned. Then fit the sensor. To do so: 4 Bajonettring vom Transmitter entfernen, Blindscheibe entfernen.
Bump Test getestet werden. Installing the Docking Station — The docking station must be installed vertically transmitter with sensor facing down in a place having low vibrations and stable temperature — close to a possible leak.
If using the Polytron 2 Remove the 2-pin connecting terminal. Keep it in a safe place and refit it when installation is complete. If using the Polytron : 3 Remove the 4-pin connecting terminal.
Attention: Spacers e. Cut excess wires short or 2 Fasten in 4-pin terminal. If installing the transmitter in areas not exposed to explosion hazard: Caution: Ex-Schutz Kennzeichnung vom Transmitter entfernen. Remove approval label from transmitter. Cut excess wires short or 2 Fasten in the middle terminals. If installing the transmitter in areas not exposed to explosion hazard: Non-Ex-hazard area Safety barrier Caution: Ex-Schutz Kennzeichnung vom Transmitter entfernen.
Slide the terminal block back into the holder. Technical data Cable type Min. Secure the lower screw and then retighten the upper screw. For the sensor assembly: 1 Slightly loosen the screws at the back of the angle support.
To insert the sensor into the extension: 3 Unscrew the bayonet ring on the end of the sensor extension. Then with light presure push upwards until the plug engages.
Intended use Remote Sensor — 83 17 — For the external installation of a sensor at a measuring point located at a distance of up to 30 m from the installation site of the Polytron Transmitter. An extension of the cable or replacement by a different cable is not permitted. Wall mounting 1 Drill the fixing holes for the mounting plate drilling template: see the pale grey illustration on page 2. Installing the sensor 3 Unscrew the bayonet ring from the transmitter and remove the dummy plate.
Note: The remote sensor is recognised automatically by the transmitter. No more installation steps are required. Drilling template 90 23 - MA Intended use Duct adapter for remote sensor — 83 17 — For mounting a remote sensor on a pipe or duct, — for gas measurement in the pipe or duct.
Drilling template: see below. Caution: To avoid faulty measurements pay close attention to the fitting of the sensor in the sealing sleeve! We recommend that the relay module should be installed by specially trained technical personnel. Intended use Relay module — 83 17 — For locally switching actuators, alarm emitters etc. Caution: use only an Allen key 5 mm without golfball.
Install the Relay Module in measuring unit 3 Bend the snap-hooks on the sides of the measuring unit cover gently outwards to release the cover. The relay cap can be removed for easier handling. Inserting the measuring unit with relay module 6 Push the measuring unit with relay module into the docking station and lower it as described in Installation Instructions 90 23 T — Order No. We recommend that the pump module should be installed by specially trained technical personnel.
Caution: Bei dem Einbau des Pumpenmoduls muss das Polytron von der Spannungsversorgung getrennt sein. Push the measuring unit up to about half height approx. The glass tubes of the pump can be inserted in these holes.
Punch or drill the holes all the way through the docking station from inside. Each hole should have a 6 mm channel. Then deburr the holes from the outside. Installing the Pump in the Measuring Unit 4 Bend the snap-hooks on the sides of the measuring unit cover gently outwards to release the cover. Position the hoses inside the housing. The seal slides over the sensor. Durch eine Reaktion bzw. We recommend that the software dongles should be installed by specially trained technical personnel.
Attention: Use only an Allen key 5 mm without golfball. Then insert the dongle in any of the three slots provided. Note: The 4pin input terminal 3 and output terminal 2 are electrically connected Make sure that the measuring unit is lowered all the way to the stop.
Short-link Link Embed. Share from cover. Share from page:. More magazines by this user. Close Flag as Inappropriate. You have already flagged this document. Thank you, for helping us keep this platform clean. The editors will have a look at it as soon as possible.
Delete template? Cancel Delete. Cancel Overwrite Save. Don't wait! Try Yumpu. Start using Yumpu now! Resources Blog Product changes Videos Magazines. Integrations Wordpress Zapier Dropbox.
Dräger Polytron® 7000
The big graphic display shows status information with the help of icons and plain text, and guides the user through calibration and configuration. The modular design allows for the subsequent retrofit onto any of the mentioned interfaces. With the Polytron remote sensor adaptor and lead it is possible to mount the sensor up to 30 metres from the polytron transmitter. This enables the user to read and operate the transmitter from a safe area, or detect toxic gasses and oxygen levels in difficult to reach areas, whilst still being able to view and configure the Polytron from a convenient location. A duct mount kit enables the remote sensor to be mounted direct in elevated pipe work. With two gas alarm relays, and one fault alarm relay, the transmitter can either be used as a stand alone gas detector with no need for a controller; or alarm devices can be activated locally saving costs in cabling to and from a controller. The analogue signal can still be transmitted back to a controller if desired.
Dräger Polytron 7000 Measuring Unit, Installation Instructions
The large graphical display usesicons and plain text to show the status ofthe instrument, and guides the userthrough calibration and configuration. Offering the advantage of reducedwiring cost and enabling the remote diag-nosis and predictive maintenance func-tionality for a reliable and flexible commu-nication network. Specificallydesigned for the demanding requirementsof a 24 hours a day, days a year sta-tionery gas detection system, the largerDrgerSensor are renown for their longlife times and superior performance. Theembedded sensor memory contains allrelevant sensor data. The Online Industrial Exhibition. Related Searches Pressure regulator Thermal protection clothing Portable analyzer Chemical protection clothing Filtration unit Mechanical protection clothing Anti-static clothing Digital analyzer Chemical protection coveralls Carbon dioxide analyzer Gas leak detector Portable monitor Compact detector Gas filter Carbon monoxide analyzer Zippered coveralls Wear-resistant clothing Thermal protection coveralls Monitoring detector Industrial tester.
Dräger Polytron 7000 Instructions For Use Manual
Docking Station Part No. Keep it in a safe place and refit it when installation. Cut excess wires short. Ex-Schutz Kennzeichnung vom Transmitter entfernen. Cut excess wires short or.